In a Gage R&R study, the number of distinct categories represents the number of data groups that a measurement system is able to distinguish in the study. If the number of distinct categories is too small (less than 5^{1}), then the measurement system may not have the resolution needed to make meaningful distinctions between parts.

The formula for distinct categories is^{1},

ndc = 1.41(PV/GRR), where

PV is the part variation from the study and GRR is the calculated Gage R&R result.

There are two common solutions for addressing a low distinct categories value –

(1) Ensure that the parts selected for the Gage R&R study represent the full spectrum of variation observed in the process.

(2) Find a gage with at least 10x resolution versus the feature tolerance. For example, a diameter with a 0.1″ tolerance should be measured with a gage that has measurement increments of 0.01″ or less.

^{1} See AIAG, *Measurement System Analysis, Third Edition*, Page 117

**Disclaimer**

Information and downloads on this website are for general educational purposes only, provided as-is, to be used at your own risk, with no warranties of any kind. While we strive to develop highly relevant content, the information and files provided on this site are not guaranteed to be completely relevant or up to date.